In the majority of CCD applications, light reaches the CCD through a lens- or mirror-based optical system. However, in some situations it is advantageous to use an image-preserving fiberoptic bundle in place of conventional imaging optics. Significant gains in the amount of light collected can be achieved by directly coupling the light source to the CCD using fiberoptics. Depending on the amount of demagnification, the gain in light collected can exceed 10x that of a f/1.2 lens. Fibers can be bonded to most front-illuminated CCDs as well as to some back-illuminated devices.
The Coherent Fiber Bundle
A coherent fiber bundle is a collection of single fiberoptic strands assembled together so that the relative orientation of the individual fibers is maintained throughout the length of the bundle. The result is that any pattern of illumination incident at the input end of the bundle reemerges from the output end with the image preserved. Imaging fiber bundles can be made in a variety of shapes and sizes, with the most common having a circular cross section. Magnification can be achieved by the use of tapered fibers in the bundle.
Teledyne Photometrics' Fiber-Bonding Process
In order to successfully couple light from an imaging fiber bundle to the CCD, the CCD and fiber bundle must be in very close proximity. Light emerges from the individual fibers at large angles, and a gap between fiber and CCD will lead to a loss in resolution. Teledyne Photometrics uses a proprietary bonding process to minimize the distance without sacrificing CCD performance. This process directly bonds the fiber to the CCD without oil layers or the use of intermediate fiber stubs that introduce losses in spatial resolution and transmission efficiency. In addition, the bond is stable and will survive the repeated thermal cycling that occurs in high-performance CCD cameras. Teledyne Photometrics' continuous innovation in fiber bonding has extended available fiber tapers to over 145 mm in diameter, coupled fibers to the largest commercially available scientific sensors, and even mated fiber bundles to high-efficiency, back-illuminated sensors.
Efficiency vs. Magnification
Besides the transmission losses through a large piece of glass, fiberoptic bundles have a transmission loss due to changes in the fiber diameter as light traverses the bundle. When light travels down a tapered fiber, a decreasing reflectance angle results in some of the light paths exiting the fiber. This appears as a loss in "effective" numerical aperture (NA). The relative loss between fibers with different magnifications can be estimated as the ratio of their magnifications squared. The larger the fiber bundle's magnification, the greater the reduction in effective NA. Fiber bundles with a 1:1 magnification, known as "stubs", provide the highest throughput. Applications requiring the highest possible light-collection efficiency benefit most by using large CCDs to reduce the amount of demagnification required.
Limitations of Imaging with Fiberoptics
A disadvantage of fiber imaging systems is that field of view is limited by the size of available fiber bundles. Currently, the largest available fiberoptic tapered bundle is 160 mm in diameter at the large end. However, to enable imaging of even larger areas, Teledyne Photometrics can create a mosaic of fiber bundles that are connected to multiple CCDs. This assembly can either be packaged in a single camera head, or into multiple camera heads, depending upon the number of bundles in the mosaic and whether or not the bundles are tapered. A second limitation of fiberoptics is the introduction of distortion and nonuniformity of response. These defects are introduced during the fiber-manufacturing process. Because these defects are static, they can be corrected through image processing. For example, response nonuniformity can be handled in most cases by flat-field correction. Gross distortion can be corrected by appropriate scaling and warping of the image data. Shear distortion, sudden dislocation in the alignment of adjacent fibers, is more difficult to correct for, due to its discontinuous nature. Fiber defect specifications are available for customers requiring detailed information.
Teledyne Photometrics' Fiberoptic Options
Many Teledyne Photometrics products are available with imaging fiberoptics. Fiber bundles range in magnification from 1:1 fiber stubs to large 6:1 fiber tapers, and in diameters up to 160 mm. Supported CCD arrays vary from 512 x 512 pixels to 2k x 2k pixels. Fiber bundles are available with extramural absorption (EMA) fibers to improve contrast, and low-thorium glass to reduce background from radioisotopes. At a customer's request, Teledyne Photometrics will also attach scintillating fiberoptic faceplates to the front of fiberoptic tapers.